Following information in an earlier post about a oxygen regulator explosion, my friend Ted Atkins has shared this information about another, and more serious incident.
Ted writes: “Mingma Sherpa of Seven Summits earlier emailed the attached photos. While attaching a Summit regulator to a Poisk cylinder for the first time the regulator exploded.
Flames erupted and he ended up burnt. He persisted and disconnected the regulator and made the situation safe. He then asked to come and see me with the regulator. The metal case is burnt through in two points. This is different and potentially more serious that the previous accident as this was an ignition.
I quizzed Mingma extensively if it was at all possible that either item had come into contact with oil, grease or paraffin. He is adamant this could not be so. We have to trust his judgement, he is very experienced in this field.
I took advice from a Ph.D. engineering consultant who had this to say –
“I’m sure a specialist could get to the bottom of this but it might be expensive. If accidents like this happened in the UK the operation would be shut down, and the failed parts taken for analysis into the root cause. This might lead to batches being recalled if manufacturing or bottle-filling error was suspected. This has happened with medical O2 regulators in the rare cases they have failed”.
I have passed what I know to Summit Oxygen and await a response. Meanwhile please be careful. Mingma was not wearing gloves or any other safety equipment. This would have saved or reduced the burns to his hands. It could be said that he has been lucky! Until Summit get back with a definitive response please use use your best judgement. I will keep you informed of any new developments as I have them.
Please share this information. Thanks.
Further information from Ted:
“I have just had a long chat with Neil greenwood of Summit Oxygen. He informs me that there was a similar incident last year which he feels was due to a seal. This only effects an older type of Summit regulator and there should only be a small number in circulation. Neil has asked me to help to remove these from circulation. I suspect that most equipment will now be at BC or en-route.
I am happy to help to resolve any issues and advise when I get there. Summit have promised to replace these regs with new ones being shipped out from UK end of April. Asian Trekking will get them to me and I will take care of any exchange required.
I hope this eases discomfort about this situation. Please contact me if you have doubts or would like advice. Meanwhile please take the extra precautions we advised: fit cylinder to reg with the gauge down, wear goggles, wear gloves.
Topout Oxygeneering Ltd
Company Registration Number
Companies House 5401945
‘Seek out life’s problems for they are the gateway to opportunity’.